CupScan II Drug Testing Kit
CupScan II are all-in-one integrated drug testing kits that provide accurate results from a sample of urine within just a few minutes. CupScan II drug test kit is a complete drug and alcohol testing kit, perfect for drug testing in the workplace where compliance with standards is paramount, or environments where maximum control is required.
This integrated cup combines the collection cup and drug test for the six most common drugs of abuse at the Australian standard AS/NZS 4380:2008 cut-off levels, alcohol and two adulterants (pH and creatinine) allowing the validity of the specimen to be confirmed.
The size and shape of the CupScan II makes it ideal for both women and men to use. Following sample collection the administrator may initiate the drug test using the supplied activator key, maintaining full control over the test conditions. The CupScan II is fast and works in minutes whilst the flat surface on the side of the cup makes the analysis and documentation of results both fast and easy. A peel away label keeps the test results out of sight until they are ready to be viewed. The CupScan II is easy to use and has been designed to minimise operator contact with the urine sample whilst continually assuring the integrity of sample and results.
CupScan II measures the following drugs and/or their metabolites at cut-off levels defined by the Australian standard AS/NZS 4380:2008, which describes procedures for the collection, detection and quantitation of drugs of abuse in urine.
|Code||Test Component||Cut-Off Level|
|COC||Cocaine Metabolites||300 µg/L|
|THC||Cannabis Metabolites||50 µg/L|
If you would like further information about these narcotics please view our drugs of abuse information page.
CupScan II is comprised of individual test strips, which undertake analysis for a specific component based on the principle of solid-phase immunoassay technology. Using this method a specific immunochemical reaction between antibodies and antigen is employed to analyse a corresponding compound within a biological matix.
The assay is based on competition between the drug conjugate and any free drug to bind with an antibody. If a drug is present in the specimen undergoing analysis, it competes with drug conjugate for antibody-dye conjugate, which is in limited supply. If the quantity of drug is equal to, or greater than, the cut-off levels of any of the drugs, it prevents the binding of drug conjugate to the antibody.
As illustrated below, this is why no band is visible on the test line zone when a result is positive, whilst the presence of a coloured band indicates a negative result.
Each CupScan II drug test kit contains the following inclusions:
- A built in test cassette
- Activator key
- Integrated Quality Control (QC)
- Results cover
- Leak-proof lid
- Integrated temperature strip
- Tamper evident security seal
- Product insert
The following instructions provide details on how to use CupScan II to analyse a urine sample for drugs of abuse and adulterants.
- Remove the CupScan II from the sealed foil pouch.
- Remove the lid and collect the sample, ensuring that the sample is above the minimum fill line.
- Secure the lid tightly and place the CupScan II on a flat surface.
- Remove the key from the lid and insert it in the side chamber, push to release the sample into the test zone.
- Place the CupScan II on a flat surface.
- Remove the result window cover to view the results. Results should be viewed at 5-8 minutes.
- Do not interpret the results after 10 minutes.
Note: If the urine sample has been refrigerated, bring to room temperature (15°C–28°C) prior to activating the test. Do not interpret the tests results after 10 minutes.
The presence of a line next to both the Control “C” and Test “T” indicates a negative result. Please note that this does not necessarily mean there is a complete absence of drug in the specimen, only that the level is below the cut-off given in Australian standard AS/NZS 4308:2008.
The absence of a line next to the Test “T” and the appearance of a line next to the Control “C” indicates a positive test result. As with a negative result, this is an indication the concentration of tested drug(s) in the specimen is above the particular cut-off level of that component.
If there is an absence of a line in the Control “C” area following the alloted time the test should be considered invalid. In this case the sample should be re-analysed with a new test, in strict accordance with the documented procedure.
Note: It is important to understand that the colour intensity of Test “T” lines can vary and that the presence of any line, no matter how faint, is still indicitive of what the test suggests.
The following documents related to CupScan II are available for viewing and download: